The model R720 color reflection densitometer utilizes the most advanced technology to ensure great accuracy, inter-instrument, and repeatablity. This instrument combines functionality and simple user interface in a small, lightweight design.
The model R720 measures: Density, Density Difference, Dot Area, Dot Gain, and Auto Function.
- Simple user interface with self-guiding prompts
- Menu-driven display guides the operator through calibration, set-up, and all measurement functions
- Extremely reliable and requiring little or no maintenance
- Multiple language user interface supported (English, Spanish, Portuguese, French, Italian, German, and Japanese)
- Easy and low cost maintenance
- Manufactured in Japan under ISO 9001 certification
- Upgradeable to the model R730
- Operation Manual
- Calibration Standard
- AC Adapter (9V, 500mA, center-negative)
- Carrying Case
- Serial Interface Cable (Optional)
- Cable Harness (Optional)
- To increase measurement functions, upgrade to the R730 for a nominal fee.
Features / Measurements
Densitometers do not actually measure ink thickness, but rather measure reflected light. A logarithmic equation is used to calculate density from the light absorbed by the surface. To calculate the absorbed light on a surface, the difference between the light reflected back from the surface and the light supplied by the densitometer is used.
Densitometers calculate density by using this equation:
Density Equation = log10 1/R, where R= % Reflection
As you can see the equation is logarithmic, which converts into the values shown on Table 1. To explain the Table further, it can be understood as a density of 1.00 is reflecting 10% of the light.
Your density reading is a guide to keeping your ink level monitored and consistent. In doing this throughout your press run you can detect any changes that could alter your print quality and allow to provide quality control over maximizing the use of your inks.
Below, are some typical wet ink densities, to follow as guidelines.
|NON-HEATSET WEB, Newspaper
Table 2: GATF-Measurements were made using a Status-T densitometer, and the density of the paper was included in the measurement.
Most densitometers give you the option to include or not include the density of your substrate into your reading. If you chose to include the measured density of the substrate or base along with the ink sample, this is known as Absolute Density. If you subtract the density of the substrate from the density of the ink sample, this is known as Relative Density.
Density Difference allows you to establish a reference density and compare this to a sample density. Say you choose your reference color and it reads Yellow 1.05D. Now, take a reading of your sample color- say it reads Yellow 1.02D. The densitometer will subtract your reference color from your sample to show the difference between the two densities. In this example, your Sample yellow deviates (-0.03D) from your Reference yellow.
Many times the naked eye is not enough. Two colors may appear to be identical when in reality they are not. You may notice changes in your image, yet you feel your density value has remained constant. The density difference function helps aid in these situations.
Keeping your dot size consistent in printing is very important. If your dot size changes throughout your press run it can have drastic effect on your image. For this reason, most densitometers provide you with dot area. This is an estimated percentage of the dot size on your print.
Changes in dot size can be the most common problem to incorrect color. In using the dot area function, the densitometer allows you to keep a monitor on your dot size throughout your press run so you can keep your color consistent.
Dot gain occurs when wet ink comes in contact with paper. As the ink dots are applied to the paper, the wet ink spreads, causing the dots to increase in size and appear darker, the size of the dot changes. This occurs in any printing job, however, the goal is to keep your dot gain consistent. For example, if the original had a dot area of 50% on the film and the resulting printed dot area was 68%, the dot gain would be 18%. Generally, the higher the screen ruling, the higher the dot gain. The densitometer will compute percent dot area and then the coverage of the printed image, thus comparing these readings to display dot gain.
As with dot area, almost any job allows for dot gain, however, if this number fluctuates or changes in any way during your press run, it can cause colors to change.
The table below shows some typical Dot Gain values for three common printing conditions.
|NON-HEATSET WEB, Newspaper
Table: GATF-Measurements were made using a Status-T densitometer, and dot gain was calculated using the Murray-Davies Equation, measures at the 50% film dot area.
The auto function allows the user to measure density, dot gain and ink trap interchangeably. In this mode, the instrument can determine if the measured target is a solid, half tone or overprint. The corresponding function will be displayed. This mode is useful when the operator uses these three functions frequently.
Measuring Geometry: 0°/45° (ANSI PH2.17, ISO 5/4, DIN 16536)
Filter Response Status: Status T, Status E, Status A
Repeatability: ±0.01D (1%)
Measuring Range: Density 0.00D - 2.50D, Dot Area 0% - 100%
Accuracy: ±0.02D (2%)
Light Source: Halogen Lamp, Approx. 2856°K
Aperture Diameter: 3.0mm (1.7mm optional)
Detector: GaAsP Photodiode
Display: 128 x 64 Dot Graphic LCD
Power Supply: Ni-Cad (4.8V), 800mAh
Recharge Time: Approximately 1.5 hours
Measurements per Charge: Approximately 4,000 (Internal Testing)
Warm Up Time: None
Measuring Time: Approximately 1 second
Operating Temp. Range: 41°F ~ 104°F (5°C ~ 40°C)
Polarization Filter: (Optional)
Security User Code: (Optional)
Dimensions: 2 3/4" W x 2" H x 8 1/4" L (72mm x 50mm x 210mm)
Weight: 1.2 lbs. (with batteries) (530 g)
Computer Output: Configurable RS-232C Serial Interface
Comes standard with a US 2-Pin flat plug.
Also available with a Euro 2-pin round plug or a British 3-pin plug.
E-mail for information and a quote.
If ordering a NiCd Battery Pack please include the below information to verify correct battery is being ordered.
- E-mail serial number of existing densitometer, or
- E-mail part number from existing battery pack, or if it shows R1, R2, or R3, or
- E-mail a picture of existing battery connector
- Models: EQP-R720-TNXU, EQP-R720-TSNU, EQP-R720-TNPU, EQP-R720-ENXU, EQP-R720-ESNU, EQP-R720-ENPU, EQP-SFTW-008, EQP-BATT-005, EQP-BATT-003 & EQP-BATT-007